The Radiation Alert® Sentry EC is a Personal Alarming Dosimeter / Ratemeter designed to ensure the safety of personnel that work in occupations with potential x-ray or gamma exposure.
The pocket sized unit has an energy compensated tube for a linear response to gamma and built in memory for recording data points for tracking accumulated exposure. The SentryCom Software option enables you to generate incident reconstruction for analysis. The software also permits you to easily set the vibrating and audio alert to your desired levels for dose (>1.0 mR/10 µSv) and dose rate (>1.0 mR/hr/10 µSv/hr).
Use the audio switch to choose between audible clicks with each count recieved or a discreet silent mode.
DetectorEnergy Compensated GM Tube
Dose Rate LinearityBetter than ±15% up to 15 R
Energy ResponseDown to 20 KeV
AccuracyTypically ±15% of reading (Cs137)
Gamma Sensitivity1.5 cps/mR/hr (Co60)
Operating RangeDose Rate: 0.1 - 15 R/hr (1 µSv/hr - .15 Sv/hr)
Accumulated Dose: 0.1 - 65 R (1 µSv - .65 Sv)
Alarm ThresholdsDefault: Dose 500 mR / Rate 50 mR/hr
Sievert Option: Dose 5000 μSv / Rate 500 μSv/hr Alarm and warning levels are user selectable with the Free SentryCom Software.
Switch FunctionsPower, Vibrate & Audible Alert, Silent Vibrating Only Alert, and Audio Clicks On/Off.
Power Requirements9-Volt Alkaline Battery(Included) Approx. 1500 hrs at normal background.
EnclosureAnodized Aluminum Housing w/ Metal Belt Clip
Temperature Range-20° - + 50°C (-4° - +122°F)
Weight243 g (8.6 oz)
Size106 x 67 x 28 mm (4.2 X 2.7 X 1.1 in)
1 year limited
Quick Start Video Guide | SentryCom Software
Frequently Asked Questions about TheSentry EC
How many data points can the Sentry collect?
The Sentry can collect up to 1024 data points. When the limit is reached, the earliest data points then are over written by the most recent data points one point at a time. How frequently each point is collected can be set with the Optional Sentry Software.
When I connect the USB-Serial Cable to the computer, why does Windows does not recognize that the device is connected?
Most Windows operating systems will download the necessary driver for the cable from Windows Update. In the event that the driver does not install automatically, you can download and install the driver for your OS at the following page by Clicking Here